Perform automatic and accurate calibration and verification of IEPE and charge mode piezoelectric accelerometers or transducers in your own lab. This calibration verification software (VR9106) provides an easy interface to calculate accelerometer sensitivity.
With the VR9106 software, accurately calibrate and verify an accelerometer to determine its sensitivity at various frequencies of interest. VR uses the most convenient and least expensive technique (per ISO 16063-21): the back-to-back comparison method, approved by the International Society of Automation (ISA).
VR’s accelerometer calibration software performs automatic and accurate calibration and verification of IEPE accelerometers and charge mode piezoelectric accelerometers/transducers. The module has a user-friendly interface to easily calculate accelerometer sensitivity. The user can perform a sine sweep controlled on a reference accelerometer to produce a calibration report and calculate the accelerometer sensitivity at the defined frequency.
Why Perform Sensor Calibration?
Early detection of incorrect sensor sensitivity
Discover sensor defects
A higher level of confidence
Greater trust in your test results
VR knows your time is valuable. Our accelerometer calibration package eliminates the need to pack up each accelerometer for shipment back to the lab to be calibrated. Our calibration software will reduce your downtime.
Custom reporting using Microsoft® Word
Back-to-back reference sensor
Reads and re-writes to TEDS sensors
Compare sensitivities and save new values
Test profile included
The VR calibration package can be used on all your accelerometers for one price, time and time again. Simply hook up each unit for calibration, run the 10-minute verification procedure, retrieve the report, and continue on with your day.
This package will allow the user to perform a sine sweep, control on a reference accelerometer, and produce a calibration report suitable for calibration record keeping. The software automatically calculates the accelerometer sensitivity at the chosen frequency. Requires SineVIEW (VR9100), a vibration controller (minimum of 2 channels), and a reference accelerometer.
Accelerometer Calibration Webinar
Accelerometer (Sensor) Calibration
Use the standard report template or customize your report using Microsoft Word. VR offers one of the most flexible custom reporting packages anywhere.
Already have one of the following? Ask for a quote today to complete your system! Vibration Research software is compatible with your existing components and we can create a custom package to complete your system.